Resolution Problems Loss of Resolution Increase in peak width
{"Image DocumentName": "Resolution Problems Loss of Resolution Increase in peak width", "Image ImageDescription": "Possible Cause. Solution. Comments. Change in carrier gas velocity. Check carrier gas velocity. A change in retention time also occurs. Column contamination. Trim the column. Solvent-rise the column. Remove \u00bd - 1 meter from the front of the column. Only for bonded and cross-linked phase. Inlet liner contamination. Clean or replace liner. Change in the injector. Check the injector settings. Typical areas: split ratio, liner, temperature, injection volume. Change in the sample concentration. Try a different sample concentration. Peak width increases at higher concentration. Improper solvent effect lack of focusing. Lower oven temperature. Choose different solvent for better solvent/sample/phase polarity match. Use a retention gap. For splitless injection.", "Image XResolution": "37", "Image YResolution": "37", "Image ResolutionUnit": "Pixels/Centimeter", "Image Software": "SlidePlayer", "Image YCbCrPositioning": "Centered"}