20.2.7.6 Additional Wiring and Equipment Requirements inside Class B Chambers.
{"Image DocumentName": "Additional Wiring and Equipment Requirements inside Class B Chambers.", "Image ImageDescription": "The requirements in shall apply to Class B chambers whether they are pressurized with oxygen or with air Electrical equipment inside Class B chambers shall be restricted to communication functions and patient physiological monitoring leads Circuits shall be designed to limit the electrical energy to wire leads into the chamber under normal or fault conditions to no more than 28 V and 0.5 W Communication wires shall be protected from physical damage and from coming into contact with flammable materials in the chamber by barriers or conduit Patient monitoring leads shall be part of approved electromedical apparatus meeting the requirements in", "Image XResolution": "37", "Image YResolution": "37", "Image ResolutionUnit": "Pixels/Centimeter", "Image Software": "SlidePlayer", "Image YCbCrPositioning": "Centered"}